AURA Touchscreen Autoclave – Class B & S

2025-10-07T15:33:21+11:00

Vendor: AURA
Type: Sterilizers
Price: 6815.00 - 7150.00 (2 variants)

The AURA Touchscreen Autoclave sets a new benchmark for benchtop sterilization. Featuring cutting-edge technology and high-quality European components, it delivers reliable, consistent results across various clinical environments.

An autoclave achieves sterilization by applying pressurised moist heat to destroy microorganisms such as bacteria, viruses, fungi, and spores. By raising the temperature above water’s boiling point (121–134°C) in a sealed chamber, it uses steam and pressure to denature vital microbial proteins, ensuring thorough decontamination. These devices are crucial in medical, laboratory, and pharmaceutical settings for sterilizing instruments, glassware, liquids, and more.

AURA Touchscreen Autoclave details:

Available in 17Ltr & 22Ltr sizes using reliable European components

Sleek Finish

Easy To Operate User Interface

Touch Screen & Buttons Allowing Fast Operation

Inbuilt Printer & USB Functions

Side mounted water outlets.

Microprocessor Control System For Precise Sensor Management

Automated Cleaning Program & Maintenance Reminders

Automatic RO & Manual Fill Options

S & B Class Cycles

Delayed Start Function

Automatic Component Diagnosis Tool

Water Quality Sensor

Five Selectable Languages

AURA Autoclave Specification:

External Dimensions: 625 x 440 x 475mm

Chamber Dimensions: 249 x 355mm (17L) / 249 x 450mm (22L)

Weight: 52KG (17L) / 55KG (22L)

Selectable 134°c Cycles: Universal Class B, Fast Class S, Prion, Custom

Selectable 121°c Cycles: Universal Class B, Fast Class S, Custom

Test Cycles: B&D Helix, Leak & Vacuum

Jacket Heater With Six Different Settings

Drying Settings Available: Normal, Extra Dry 5 min, Extra Dry 10 min

Included Accessories: Tray Rack, 3 X Trays, 2 X Drain Hoses & Connectors, Tray

Handle, Door Seal, Chamber & Water Tank Filter Mesh, Paper Roll, USB Drive,

AC Power Cord

Required Voltage: 120 – 230V AC

Rated Frequency: 50 / 60 Hz

Rated Power: 1800 VA

Sterilising Temperature / Pressure: 121°c @ 100-130 Kpa, 134°c @ 200-240 Kpa

Water Tank Capacity: 3.5L

Operating Temperature: +5°c – +40°c

Relative Humidity Range: 30% – 75%

Warranty: 2 Years

Certifications: CE, ISO13485-2016, ASME / Worksafe, EN13060

ORDER CODES:
AURA17 – 17L
AURA22 – 22L

NOTE: Professional installation & validation required. Please make sure your autoclave is serviced in accordance with the users manual and machine prompts to ensure it functions to its optimum. All servicing must be completed by our network of authorised agents to maintain machine warranty.